LIFEPAK 20 and LIFEPAK 20e External Defibrillator/Monitors by Physio-Control Inc.: Class I Recall Due To Power Supply Failure
(July 2, 2010)
Audience: Emergency Medicine, Cardiology, Risk Manager
ISSUE: A failure on the power supply assembly can result in either "No DC power" or "No DC or AC power". A failure of DC (battery) power can result in the inability to deliver defibrillation therapy if the device will not turn on using DC (battery) power and no AC (line) power is available.
BACKGROUND: The LIFEPAK 20 and LIFEPAK 20e defibrillator/monitor is designed for use by trained medical personnel in hospitals and clinic settings to monitor patient heart rhythms and to treat patients experiencing cardiac arrest. Approximately 42,943 devices were distributed worldwide between September 16, 2002 and September 27, 2007. These devices were manufactured from July 31, 2002 to September 19, 2007.
RECOMMENDATION: All affected power supplies will be updated. Customers are advised to keep the defibrillators in service and follow recommended daily Operator Checklist steps while service updates are scheduled. See Recall Notice for contact information.
THE AMERICAN SOCIETY OF ANESTHESIOLOGISTS
Founded in 1905, the American Society of Anesthesiologists (ASA) is an educational, research and scientific society with more than 52,000 members organized to raise and maintain the standards of the medical practice of anesthesiology. ASA is committed to ensuring physician anesthesiologists evaluate and supervise the medical care of patients before, during and after surgery to provide the highest quality and safest care every patient deserves.
For more information on the field of anesthesiology, visit the American Society of Anesthesiologists online at asahq.org. To learn more about the role physician anesthesiologists play in ensuring patient safety, visit asahq.org/WhenSecondsCount. Join the ANESTHESIOLOGYTM 2014 social conversation today. Like ASA on Facebook, follow ASALifeline on Twitter and use the hashtag #ANES2014.